Acidity and pH Titrator for Milk and Cheese (Hanna Instruments)
A low cost, easy to use, advanced microprocessor-based automatic titrator and pH meter that performs automatic analysis with all the necessary calculations through a clear and simple interface. Integrated built-in algorithm analyzes the shape of the pH electrode response and determines the reaction completion. By pressing the START key, the instrument automatically conducts an end point titration and result is immediately displayed. A simple and reliable peristaltic pump ensures accuracy and repeatability.
Comes with a pre-programmed analysis method designed for Total Titratable Acidity measurements on milk as well as a pH electrode, temperature probe, solutions and tubing.
The determination of total acids in dairy products is made according to a neutralization reaction; that is the reaction between the acids found in dairy products and a base. This type of reaction forms the basis of titration methods of analyzing acids. Titratable acidity is measured on a degassed sample at the end point of 8.30 pH. The results are expressed in % lactic acid (l.a.). This is the unit used in the U.S.A. and Canada. The instrument also has the capability to express °Soxlet Henkel (°SH), °Thorner (°Th), or °Dornic (°D).
About acidity measurement and its significance in the dairy industry: There are two fundamentally different conventions for expressing acidity in dairy products: titratable acidity and hydrogen ion concentration, or pH. The former expresses total acidity but does not measure the strengths of the acid, while pH indicates acid strength. Acidity affects taste, thus this parameter is tested to determine the quality of the milk product. As milk acidity increases over time, measuring this parameter is also a means of monitoring storage conditions. Acidity is determined by an end-point titration using sodium hydroxide (a base) and is defined as the consumption of base necessary to shift the pH value from 6.6 +0.1 (corresponding to fresh milk) to a determined basic pH value. While pH 7.0 is the actual point of neutralization, phenolphthalein is commonly employed as a color indicator to determine the end-point of reaction and with it, color change occurs at pH 8.3. Titratable acidity is expressed percent lactic acid, the use of which reflects the titration method and strength of base employed during titration.
Makes measuring both pH and acidity of dairy products possible within one compact unit. The titration method is a potentiometric end-point determination at a pre-determined pH value. This titrator eliminates the subjective end-point color-change detection determined by the human eye, and instead employs the sensitivity and accuracy of a pH sensor, virtually eliminating error. After performing a pump calibration with the supplied standard, you can then make titrations, expressed in the desired unit, using the same titrant. This eliminates the inconvenience of changing tubes, purging the titrant for tube cleaning and being sure that you have the right titrant concentration thus eliminating wasted time and titrant. The quantity of sample needed is much smaller in comparison with the classical method, where 100 mL of product is used.